Algorithm for Estimation of Measurement Devices Software’s Metrological Reliability
Nowadays, parameters, characteristics and functionality of measurement devices basically depend on microprocessors software. Software used in measurement devices overtakes functions from a hardware part of the device. For example, data processing algorithms, auto diagnostic control functions are implemented using software. Basically during verification of the device, just a hardware part of the device is verified. But devices metrological reliability depends on implemented auto diagnostic algorithm. In this article algorithm on how to estimate metrological reliability of measurements devices software is presented. Few experimental testing results of heat meters autodiagnostic algorithm software presented. Ill. 4, bibl. 6, tabl. 1 (in English; abstracts in English, Russian and Lithuanian).
How to Cite
The copyright for the paper in this journal is retained by the author(s) with the first publication right granted to the journal. The authors agree to the Creative Commons Attribution 4.0 (CC BY 4.0) agreement under which the paper in the Journal is licensed.
By virtue of their appearance in this open access journal, papers are free to use with proper attribution in educational and other non-commercial settings with an acknowledgement of the initial publication in the journal.